ANSTO Publications Online >
Conference Publications >
Conference Publications >
Please use this identifier to cite or link to this item:
|Title: ||A new method for separation and purification of beryllium and aluminium|
|Authors: ||Child, D|
|Issue Date: ||24-Mar-2011|
|Publisher: ||12th International Conference on Accelerator Mass Spectrometry (AMS-12)|
|Citation: ||Child, D., Simon, K., (2011). A new method for separation and purification of beryllium and aluminium. 12th International Conference on Accelerator Mass Spectrometry (AMS 12), 20th - 25th March 2011. Museum of New Zealand: Te Papa Tongarewa, Wellington, New Zealand.|
|Abstract: ||The DIPEX® resin manufactured by Eichrom was originally developed for the separation and preconcentration
of actinides from nuclear industry waste streams and digested environmental
samples. The resin has in recent years been investigated for use in separation of beryllium from a
wide range of other elements which interfere with the determination of beryllium by ICP-AES.
Application of this resin to purification of beryllium from environmental samples for analysis by
Accelerator Mass Spectrometry (AMS) may allow for simplified sample processing in a single step
and reduced interferences.
AMS measurements of the cosmogenic isotope 10Be however are susceptible to a different selection
of interfering elements than those which affect ICP-AES measurements. 10Be measurements are
most commonly performed on melted ice and purified quartz. Both these sample types are
generally low in elemental impurities such as magnesium, calcium, aluminium, titanium and iron
with concentrations typically no more than a few hundred ppm in purified quartz and only a few
ppm in ice however potential interferences still exist that must be removed prior to analysis. Two of
the most significant interfering elements are boron and titanium. The isobar 10B, if sufficiently
abundant, interferes with 10Be measurements by increasing the measurement background.
Titanium has a retarding effect on caesium sputter ionisation and so on ion source output.
The DIPEX resin has been evaluated as a single column purification and separation method for
preparation of samples for AMS analysis. In this paper we present data on this evaluation
• Evaluation of the resin capacity and selectivity for aluminium in a simulated matrix
representing digested quartz and melted ice
• Evaluation of the selectivity of the resin for beryllium in a simulated matrix representing
digested quartz and melted ice
• Assessment of the resin’s capacity to separate interfering matrix elements from the beryllium
fraction.Copyright (c) 2011 AMS12|
|Appears in Collections:||Conference Publications|
Files in This Item:
There are no files associated with this item.
Items in APO are protected by copyright, with all rights reserved, unless otherwise indicated.